Skip to Main Content
| Create Account
| Sign In
IEEE membership options for an individual and IEEE Xplore subscriptions for an organization offer the most affordable access to essential journal articles, conference papers, standards,
eBooks, and eLearning courses.
Learn more about:
IEEE Xplore subscriptions
Your organization might have access to this article on the publisher's site. To check,
click on this link:http://dx.doi.org/+10.1063/1.1149656
The dynamics and noise of a dc superconducting quantum interference device (SQUID) with the McCumber parameter βc=2πR2IcC/Φ0 close to the unity (where Ic, R, C are the critical current, the shunt resistance, and the capacitance of the Josephson junctions comprising the SQUID, respectively, and Φ0=2.07×10-15 Wb is the magnetic flux quantum) integrated with a planar spiral input coil have been experimentally studied. The length of the spiral input coil was chosen to match its λ/4 microwave resonance frequency to the plasma resonance frequency of the SQUID. The input coil resonance enhances the overall quality factor Q of the Josephson oscillations in the SQUID and, as a result, increases the dynamic resistance Rd and the gradient of the flux-to-voltage characteristics ∂V/∂Φ without hysteresis. This relaxes the tolerance for the βc parameter, simplifies the technological process, and improves the yield of devices. A dc SQUID with loop inductance L=31.4 pH, βc=0.72, and a six turn input coil has demonstrated a nondistorted quasisinusoidal flux-to-voltage transfer function with an exceptionally large modulation depth of approximately 140 μV peak-to-peak. A spectral density of the intrinsic magnetic flux noise as low as 3.5×10-7 Φ0/Hz1/2 has been measured in the double stage configuration at a temperature of 4.2 K using direct read-out electronics. This corresponds to the intrinsic energy resolution of Є=12.5h. In combination with an intermediary transformer, the current res- olution of the SQUID is as low as 1.25 pA/Hz1/2 with an input coil inductance of 58 nH. The coupled energy resolution is Єc=45h in the white noise region. © 1999 American Institute of Physics.
Review of Scientific Instruments
Date of Publication:
A not-for-profit organization, IEEE is the world's largest professional association for the advancement of technology.
© Copyright 2013 IEEE - All rights reserved. Use of this web site signifies your agreement to the terms and conditions.