Skip to Main Content
IEEE membership options for an individual and IEEE Xplore subscriptions for an organization offer the most affordable access to essential journal articles, conference papers, standards,
eBooks, and eLearning courses.
Learn more about:
IEEE Xplore subscriptions
Your organization might have access to this article on the publisher's site. To check,
click on this link:http://dx.doi.org/+10.1063/1.1149620
The use of a 200 MHz lock-in amplifier was demonstrated as a low cost instrument for frequency domain measurements of nanosecond fluorescence lifetimes. The lock-in directly provided both the dc bias and the ac signal used to modulate the intensity of a blue light emitting diode excitation source. The emission was measured by a photomultiplier tube and the resulting signal was sent back through a dc block to the lock-in with no external signal processing or heterodyning required. The system was highly accurate at measuring phase and modulation up to 80 MHz and moderately so up to 100 MHz. The fluorescence lifetimes of several standard fluorophores (Fluorescein, Rhodamine B, and [Ru(bpy)3]2+) were measured by the lock-in, and the results agreed closely with those made on a research grade fluorometer. The entire lock-in based system costs less than US $10,000 to build and can be controlled by any standard computer through a GPIB or serial connection. The system is also portable, consumes little power, and can easily be configured for use with fiber optics, making it ideal for use with fluorescence lifetime based sensors. © 1999 American Institute of Physics.
A not-for-profit organization, IEEE is the world's largest professional association for the advancement of technology.© Copyright 2014 IEEE - All rights reserved. Use of this web site signifies your agreement to the terms and conditions.