Skip to Main Content
IEEE membership options for an individual and IEEE Xplore subscriptions for an organization offer the most affordable access to essential journal articles, conference papers, standards,
eBooks, and eLearning courses.
Learn more about:
IEEE Xplore subscriptions
Your organization might have access to this article on the publisher's site. To check,
click on this link:http://dx.doi.org/+10.1116/1.589398
A novel high vacuum scanning force microscope (HV-SFM) using a piezoelectric cantilever and the phase detection method was developed. A self-excited force sensing PZT microcantilever with dimensions of 125×50×3.58 μm is vibrated in vacuum at its resonance frequency by applying an ac voltage to its PZT layer. The piezoelectric microcantilever possesses a spring constant of 16.7 N/m and a vibrational quality factor of 300 in air and 807 in vacuum. As we know, force gradients acting on the vibrating cantilever cause changes in vibration amplitude and phase. The change of the phase of the output piezoelectric current, which corresponds to the phase variation of cantilever vibration, is measured as the controlling signal in this method. The piezoelectric HV-SFM that was constructed has been proved to be stable and easy to handle in vacuum. Its vertical resolution is higher than 1.1 Å. High resolution images of an evaporated Au film can be obtained in vacuum by this new dynamic HV-SFM. Grains of about 5 nm can be seen clearly. The theoretical sensitivity of phase detection with piezoelectric cantilevers is also analyzed and compared with that of amplitude slope detection in this article. © 1997 American Vacuum Society.
A not-for-profit organization, IEEE is the world's largest professional association for the advancement of technology.© Copyright 2014 IEEE - All rights reserved. Use of this web site signifies your agreement to the terms and conditions.